What is the details for Pulse Output Water Meter
Product Description--Pulse Output Water Meter
It is equipped with a reed switch assembly which can be connected to remote reading system, The Reed Switch Pulse sends out electric signals per a preset water quantity.
--Magnetic drive, protected against external magnetic tampering;
--Vacuum sealed register, frost resistant, keeps clear reading for long time;
--Brass body can be painted
--External adjusting device;
--For cold water;
--Remote transmission device can be added upon request
--Many kinds of sensors; such as Reed Switch; Hall and Weagand can be supplied upon request Pulse output may be 1l, pulse 10l,pulse;
--The meters conform to ISO4064 Standard Class B
2. Working condition;
--Water pressure≤1MPa PN, 1, 6MPa; 16bar;
--ΔP≤0 1MPa; Class B;
--ΔP≤0 063MPa; R80; R100;
From minimum flow-rate qmin; inclusive to transitional flow-rate qt exclusive ±5%
From transitional flow-rate; qt; inclusive, to overload flow-rate ; qs; exclusive ±2.
4. Installation And Usage for the Pulse Output Water Meter
Select the right water meter according to nominal flow rate which is less than or next to the rate through the pipe. It is not a good way for the user to buy the water meter based on the pipe size.
The place to be installed should be free from sunshine, frost, pollutant and water. During the period of frost, the inlet Valve
in the meter should be closed and the outlet valve and water tap should be open except for being packed for thermal insulation, thus to prevent the damage from being frozen.
Meter should be installed horizontally;and the digits on the dial is up to the face and arrow direction in the meter should be same with the flow;
The stone, sand and slight impurities in the new installed pipe should be cleared off for the sake of the meter fault;
For the purpose of metering accurately, the water tap should be higher than the meter;
Water meter should be not connected with pipe and there are live interface (e.g. pipe interface, connection screw, sealing washer etc.) between the meter and pipe. The meter should not be damaged due tothe over-stress caused by pipe and its elements during installation.
1. Materials, brass body, ABS inner core, brass sealing ring, pulse output wire
2. Working Condition, Temperature ≤50°C
3. Measured by Cubic Meter
4.Quantity of Flow:
Main Technical Data For Pulse Output Water Meter
Looking for ideal Photoelectric Coding Remote Reading Water Meter Manufacturer & supplier ? We have a wide selection at great prices to help you get creative. All the Photoelectric Coding Reading Water Meter are quality guaranteed. We are China Origin Factory of Reading Water Meter. If you have any question, please feel free to contact us.